RRDE-3A Rotating Ring Disk Electrode Apparatus movies
Here are some movies to guide you with your RRDE-3A Rotating Ring Disk Electrode Apparatus.
A guidance for the setup of your RRDE-3A, also a quickly instructions for the maintenance of your system.
- RRDE-3A easy setup
- RRDE-3A disassembling/assembling for maintenance
- RRDE-3A motor rotation issue
- Spin coating
- Maintenance II
- Rotation issue
- Spin coating
RRDE-3A easy setup
Follow this movie to setup your RRDE-3A in 5 minutes.
This movie will show you the content of the system, and the setting of the gas purging and rotation, in a manual mode. For the remote mode, please follow the instruction for the manual.
RRDE-3A disassembling/assembling for maintenance
This movie will show you how to disassemble your RRDE-3A for maintenance.
By disassembling the RRDE-3A, you can Replace carbon brushes, shaft, and bearing.
RRDE-3A maintenance procedure
This movie will Introduce the maintenance procedure of RRDE-3A rotating ring disk electrode device Ver.2.0.
If the RRDE-3A has been used for a long time, dirt may accumulate on the motor shaft side, resulting in measurement noise, this video will show you how to clean the shaft.
In addition, from Ver. 2.0, if the warning of cleaning (Cleanup shaft!) is displayed on the LCD display, please carry out cleaning instruction on this video.
RRDE-3A motor rotation issue
This movie will show you how to fix a rotation issue.
If the coupling has entred inside the aluminium frame, it may block the rotation.
If carbon brushes are broken, please change them.
Check the bearing assembly and the shaft. If they are not in good condition (corrosion, twisted...), please change them.
Spin coating using RRDE-3A rotating ring disk electrode device Ver.1.2.
- For easy confirmation, we drop a large amount of water.
Download the Spin coating using RRDE-3A rotating ring disk electrode device Ver.1.2. movie